Pointscale 120/150 manual
**SPECIAL NOTICE** To turn on your Pointscale, press and quickly
release the ON key. Do not hold the on key down or the scale will
turn back off.
MANUAL & WARRANTY GUIDE
PLEASE READ COMPLETE INSTRUCTIONS BEFORE USE
Thank you for purchasing the Pointscale 150. With normal care
and proper treatment it will provide years of reliable service.
Please read all operating instructions carefully. You can visit
us at www.goodscale.com
.KEEP THE FOLLOWING POINTS IN MIND:
* Avoid lengthy exposure to extreme heat or cold, your scale works
best when operated at normal room temperature. Always allow the
unit to acclimate to a normal room temperature for at least one
hour before use.
* Allow sufficient warm up time. Turn the scale On and wait 60
seconds to give the internal components a chance to stabilize
* The cleaner the environment the better. Dust, dirt, moisture,
vibration, air currents and proximity to other electronic equipment
can all cause an adverse effect on the reliability and accuracy
of your scale.
* Handle with care. Gently apply all items to be weighed onto
tray top. Although this scale is designed to be quite durable,
try to avoid rough treatment as this may permanently damage the
* Avoid shaking, dropping or otherwise shocking the scale. This
is a precision instrument and MUST BE HANDLED WITH EXTREME CARE.
* Only operate the scale on a stable, vibration free surface.
IMPORTANT: Do not operate near cell phones, radios, computers,
or any other electronic device.
If your scale ever performs slowly or inaccurately, please replace
1) Three “AAA” size ALKALINE batteries are required.
2) To install batteries:
a) Release the battery cover by sliding the locks in-wards.
b) Place batteries into battery compartment aligned correctly.
c) Replace battery cover and lock it.
DO NOT USE EXCESSIVE FORCE !
3) The scale is now ready for battery operation.
*Remove the batteries if you plan to store the scale for longer
then 10 days.
KEY PAD FUNCTIONS
Mode selection: Press the key once to change weight unit.
Calibration: Depress this key for four seconds to begin calibration
as outlined under Calibration Procedure on the next page.
Press this key to turn unit on. Once the unit is on the same key
operates the additive memory feature. The following steps outline
the procedure for cumulative weighing of samples:
1) Place container on tray top, wait for stable reading, then
depress “TARE” to return scale to zero and permanently
remove the empty containers weight from the additive procedure.
2) Add first sample until desired weight is achieved.
3) Press ON key to reset unit to zero. The memory symbol “m”
appears in the upper left corner indicating the previous weight
value has been stored.
4) Add second sample until desired weight is achieved.
5) Press key again. The “m” symbol will disappear
and the display will show the combined weight of the first and
second samples. Repeat steps 2 through 4 for additional items.
NOTE: When the entire load is removed from the scale the display
will flash a negative value equal to the tared value of the container.
To reset the scale to zero press the tare key.
Press tare to reset the scale to zero. Tare can be used for eliminating
the weight value of a container. Turn the scale on, Place an empty
container on the scale and press TARE. The scale will reset to
0.0, then place the items to be weighed in the container and the
correct weight of the items will appear on the display.
NOTE: When al weight are removed from the weighing tray, the tared
value of a container will be displayed as a negative number. Depress
tare again to return the display to 0.0.
Press and hold to turn off the scale
.** These electronic scale are precision instruments. Do not operate
near an in-use cell phone, radio, computer or other electronic
device. These devices emit RF and can cause unstable scale readings.
If your scale ever performs poorly, try moving the scale to a
different room or location.
Important: Please turn the scale ON and allow it to warm up for
30 seconds before calibration. If Possible please calibrate the
scale on your SCALEPAD (included in the box), a mousepad or similar
surface (These pads are very stable and absorb vibrations). Repeat
calibration at regular intervals if the scale is subject to weather
or altitude changes. Incorrect calibration can occur if you do
not follow the steps exactly. If your scale does not perform accurately,
please try replacing your batteries before you calibrate.
1) Place the scale on a Flat stable surface. Turn the scale on
and allow it to stabilize for 30 seconds.
2) Press and hold the “CAL/MODE” key for 5 seconds,
CLO will briefly display on the screen and then 100.0 will appear.**(you
MUST wait for 100.0 to appear before proceeding)**
3) Gently place the 100 gram weight on the scale, wait 3 seconds,
then press the “OFF” key (only press it once).
4) Wait 10 seconds, the scale should show CLF and then return
to active weighing. Gently remove the weight. Sometimes the scale
may show a leftover value such as 0.1g, If you wish you may re-zero
the unit by pressing the tare key.
NOTE: if after calibration your scale does not read accurately,
calibration error and the calibration process should be repeated.
Please calibrate on a scalepad, mousepad or similar surface if
INACCURACY / ERROR___________________________________
The primary reasons for inaccuracy or malfunction are low batteries,
incorrect calibration, overload or operating on an unstable surface.
Please keep this in mind and maintain and operate your scale properly.
The scale is a precise instrument and must be handled with the
upmost care and caution.
* Power Up Segment Test
When first turning the unit on, all segments of the display will
appear as shown below. This display will remain for approximately
3 seconds and then reset to zero. The display may also change
brightness during this time and show a blank screen for a moment
before 0.0 appears
+ 220.127.116.11 >*
Stable Reading Indication
This is a very precise scale - the display may seem to wander
or jump when weighing. This is due to air currents or vibrations.
Stable weighing is achieved when the display remains fixed for
When an applied load exceeds the capacity. “EE” will
appear on the display. Remove excessive load immediately. The
unit should return to normal operation.
!Remember: You can permanently damage the scale by overloading
* Negative Value
Any tared value or a value left in memory will be displayed as
a negative number once all weight is removed from the unit. Depress
tare to return the unit to zero.
The memory function is designed to allow the user to store values
and to then progressively add these values into a sample or procedure.
Unlike the tare feature, the unit returns to zero and a memory
indication appears at the upper left of the display. After a subsequent
weight is added, press memory again. The original value is added
to the current value giving the total gross weight of the current
sample. Different samples are needed and a total gross weight
of all samples is required. This process is further outlined on
the previous page.
* Auto off
An auto shut off feature is provided to conserve battery power.
The unit will automatically turn off after approximately 3 minutes
This scale is warranted from all defect in materials or craftsmanship
for 5 years from the date of purchase to the original consumer
purchaser. Any defect created by misuse, abuse, overload or physical
damage is not covered. Returns must be Mailed to the return center.
We do not accept returns of used scales from our resellers. You
must return all scales directly to our repair center